Power meter & frequency counter (English)
The input board is a small PCB mounted directly on the connector.
Updated 3. Sep. 2002.
If you are from Denmark ! read the Danish version !!
In the good old days I had some parts but now
This Wattmeter is in ELEKTOR MAGAZINE OKTOBER 2002, they will sell PIC and PCB
The extra input B is for a later SWR brigde project, also the TX serial output will be in use later.
The uncalibrated signal response is: +1/-1 dB from 1MHz to 450MHz.
Input SWR vill varry from 1.00 to 1.30, depending on input frequency.
To make the SWR this good, you need to assemble the input circuit correct and adjust the capasitor.
Input power range: -60 to +30dBm that is 1 nW to 1 Watt.
This instrument can be used and calibrated from 1 kHz and up to 500 MHz
It is possible to measure power relative all the way up to 900Mhz
A software routine can calibrate the 0dBm point at 5 different frequencies to make this instrument accurate within 0.5dBm !!
The calibration data is stored in EEPROM so the instrument will remember all, also without power.
At frequencies above 300Mhz this instrument should not exceed inputs over +20dBm (100mW) to keep the good accuacy
This is a documented weekness in the AD8307, this is not a big problem, if you are aware of this, then it’s just a matter of using the right input attenuator
If battery opperation: do not connect R30, also change R26 and R27 to 4k7
Ideas taken from articles in: QST June 2001 page 38 and Funkamateur 12/99 page 1383 And Elektor 1/99 page 26
The display, shows dBm from -60 to +30dBm, RF Voltage and RF Power and Bargraph in 1db step.
The backlite is verry powerfull, I had to change R30 to 10E
Also I must complement the POWERTIP display for it’s exelent contrast and viewing angle and low prize !
I use a POWERTIP display with 20 caracters in 2 lines with LED light.
Powersupply current is:
With no light: = 30 mA. (R30 = NC)
Normal light: = 120 mA. (R30 = 10E)
Power light: = 200 mA. (R30 = 4E7)
The cabinet is homemade of glasfiber PCB, easy and cheap to construct.
There is made plenty of room for a batterypack, and also I like it not to be too small
so that it’s hard to come in and make modifications.
The NEW version of the PCB Now the input coil has been removed
The AD8307 is SMD SO-8, the extra input board is for a later project (SWR bridge), more about this next year.
The TX serial out pin is for several future projects, that uses a PC to draw curves and more.
The bottom PCB side is mostly shilding PCB size is 79mm wide and 37mm height
Here is all the measurements curves
Software information users guide and download
PC logger software for windows
Complete part list
Assembly and solder instructions
Digital Wattmeter construction-competition
Serial output format and hardware sorry this page only in Danish
Front plate and box design Here is my sugestion about how to do it
Comments, Ideas, News from all of us to all of you
Output to freq counter and DFD4 counter modifications
F6GOG Wattmeter page A must see page !!
How To Measure High Power
The digital-powermeter input range is 1nW to 1W.
To measure more power an attenuator can be mounted in front of it, but they are expensive !
A good and cheap solution is to use a -40 dB tab.
The Wattmeter readout can be set to show correct values, using five standard attenuators,
Attenuator settings: none, -10, -20, -30, -40 and -50dB, by this it is now possible to measure from 1nW up to 100 kW
Here is my simple homemade -40dB tap, using 3 BNC connectors a small cheap box and a few resistors.
The four first resistors is 620 Ohm each the bottom resistors is two 100 Ohm in parallel.
This uncompensated circuit can be used up to 170 Mhz and up to 50 Watts continius
This is a much better solution, but also a bit harder to make
This -40dB tab has good SWR up to 900 MHz, and the -40dB output signal has a perfect flat response up to 600Mhz
The 2.7mm stripline needs to be verry acurate, the PCB is double sided 1.6mm thick.
note that the stripline goes all the way as close as possible to the connector isolation
Here is a closeup, note the tab compensation wire, that is adjusted close to the two first resistors,
by this way it is possible to adjust a perfect flat -40 dB response up to 600 MHz
This tab can also only handle 50 Watts, limited by the four 1/4 watt resistors
To get -40dB a 100:1 voltage divider is made with 50 Ohm output impedance, so when this output is loaded
the total load resistance is 25 Ohm, now that we have the output bottom resistance of 25 Ohm we can calculate
the total resistance of the 100:1 voltage divider, 100 * 25 = 2500 Ohm so the top resistance is 2500 – 25 = 2475 Ohm
if four resistors is used they shuld be 2475 / 4 = 618.75 Ohm so I use 620 Ohm that is a standard
Power in the resistors: I use 250 mW resistors * 4 = 1 Watt the power in the bottom resistors is negible..
To calgulate the MAX RF power level, U = square root P * R, it was 1W and 2500 Ohm this give 50 Volt, this is equal to 50 W
the resistors can stand the double power for about 1 minute and 4 times a few sec, so to my needs this is fine
a 30dB attenuator
Ready made attenuator modules is made by several companies. Here is a 30 dB 100Watt from RF Power
Their web site www.rf-power.com
In this case it can not with stand 100 Watt so long, due to the lack of heat sink
The 30 dB attenuation is verry acurate from 0 to 500 Mhz, it has also perfect return loss
I know a nice person that have access to a CNC machine, he made this alu box for me.[—分页—]
Front Plate and case creation
The front plate, width = 204 mm Hight = 54 mm
I have made this front plate design with my PCB program
Then it is easy to make the rest few plates and solder them to a box.
After that, paint it black or what ever collor you like.